What settings are required in the configuration when connecting a 500 Series to Toolcrib?
To connect a 500 in Toolcrib you will need to setup the software with the following:
After lauchingToolCrib, you are presented with the list of instruments that you can connect to, Select the 500.
Then Instrument Connection Parameters will open.
Protocol should be set to Auto.
Serial Port should be set to Auto.
BaudRate should be set to 9600.
Then select Connect.
If connection fails make sure that the 500 properly connected to the computer. The valid Communication Ports for connection are 1-4, if the unit is not connected to one of these Communication Ports it will not Connect.