Note that all X-Rite 400's, 900's, SP's, MA's, DTP, and Photo scanning densitometers use an RJ45 (type) female interface connector. While identical in dimension to a true RJ45, the interface in X-Rite's products is a 10 circuit design.The interface cable described below allows X-Rite 891/892 densitometers to be directly attached to many Hayes type modems. Specifically the modem must be equipped with a DB25s (female) connector. In addition it must be electrically equivalent to a "standard" Hayes (DCE) interface (see Related SUpport).
|RJ-10c Pin#||Function||DB-25p Pin#||Function|
|5||HS_IN "A"||22||Ring Indicator|
|9||Digital Ground||7||Signal Ground|
|10||HS_OUT "B"||4||Request to Send|
|6 and 20||Internally jumpered|
|Shield||drain wire||Shield||drain wire|