962 Firmware - F203
Download: 962 0°/45° Instrument Firmware
Previous Version: D411
Release Date: 2/9/2005
File Type: PC - Self-extracting Executable
Download: PC Version
Notes: What's new in the 939/963/964 Build vF203 release :
Fixed a "skip next tag" bug in job mode -- If the customer created a job using 'next' tag(s) assigned to a particular measurement, and the customer decided to skip that particular measurement, the tag(s) assigned would not get cleared out properly. So the next measurement taken would end up with the tag(s) from the skipped measurement in addition to its own tag(s). More erroneous tags were assigned to the next measurement taken depending on how many measurements were skipped. The fix was to remove next tags whenever the 'next' button is pressed in Job mode which skips to the next stopping point in the job.
Lowered the maximum temperatures allowed for fast charge to operate from 44C to 42C.-- Over the past year there have been problems with the instrument when used in a 100% tethered mode. Occasionally an instrument would end up getting the "batteries are overcharged" message. The message shows up when the thermal cutoff in the battery pack opens up when it gets too hot. The cutoff is designed for 50C but has a 10% tolerance which may end up being around 45C worst case. Originally cutoff for Fast Charge was set at 44C. It has been noticed that even after Fast Charge is turned off at 44C battery temp. can continue to climb another 1.5 degrees to 45.5C. If that happened battery temp. would rise above the 45C worst case cut off an potentially cause the "batteries are overcharged" message. With the Fast Charge cut off set at 42C, the temp rise after cut off continues to about 43.5c and does not trip the thermal cutoff.
Added the ability to upload and download Japanese Characters (Katakana and Hiragana) to and from X-RiteColor Master software (standard & project names, job text, etc)
Updated code so that the reflectance graph shows both Sample and Standard curves.