Density value displayed seems too low when reading in dot mode
Density value displayed seems too low when reading in dot mode.When reading dot area or dot gain, a density value is displayed after reading the solid patch. The value shown is density minus paper. If you typically measure density in the absolute mode you may notice the value displayed when in the dot function is lower than expected. There is nothing wrong, it is simply the result of subtracting paper desity from solid ink density.
Questions? Need a Quote? Contact Us. 1.800.248.9748