Open the ShadeVision program (with the instrument docked), select "Advanced Menu", and then "Calibrate Now". After the instrument is calibrated, capture a tooth measurement and dock the instrument. Cancel the measurement upload (click on the red X) and then select "Instrument Health" from the "Advanced Menu". Adjust the contrast slider tab in the upper left corner of the "Instrument Health" window until the tooth image on the docked instrument is acceptable. Select "Save" to retain this new setting. Selecting "Default" will return the LCD to the factory setting. Click the green check at the bottom on the window to save your changes.
Questions? Need a Quote? Contact Us. 1.888.800.9580