X-RiteColor Master Web Edition - Communication error with 530
Problem: Trouble connecting 530 to web edition. Does take a reading, but will not pass info to software. Server is installed in VA. Used to work when on local machine in Oklahoma
Solution: Re-set the instrument and then changed the instrument handshake from RCI to ICP setting.
Questions? Need a Quote? Contact Us. 1.800.248.9748