Users may need to know how to configure the 390/391 to communicate with X-Read software
x-Read for Windows
Medical Cost Containment Consultants
PC (Win3.xx, WIN95, WIN98)
Software purchased after August of 2002 may use SE108-92 (cable) and the SE108-99 (coaxial to phone jack - power supply adapter). Older versions will likely require the more complex cabling listed below:
Customer will need either the SE108-09 ("Y" cable) or SE108-69 (cable).
If the PC is equipped with a DB25 type interface port the use an 881-71 (DB25s female) adaptor.
If the PC is equipped with a DB9 type interface port then use either the 881-71 (DB25s female) adaptor AND an SE08-232 (25 to 9 converter) OR simply use an 881-91 (DB9s female) adaptor.
An old player with a new name (company and software). Please note that x-Read for Windows replaces the previous DOS programs M-Read and X-Read.
Setup is fully described in the x-Read operation manual. First load Epson I/O preset *. Next return to the configuration menu and set the following:
Versions v6313 and higher have an I/O preset designed to quickly pre-configure the instrument. The notes above will work in any case... but if you have the X-Read preset you can save a few keystrokes: Use X-Read rather than Epson where possible.
If the SE108-09 "Y" cable is used, the "A" end of the cable must attach to the PC. The "B" end reserved for modem communications only.
The support number for Medical Cost Containment Consultants (Dave Norby) is: 608-846-9811 begin_of_the_skype_highlightingFREE
Questions? Need a Quote? Contact Us. 1.800.248.9748