|Notes :||ACR type database storage and reporting for mammography processor control. Setup is fully described in the X-Read operation manual. In brief, Per Dave's book pgs 9..10; first load Epson I/O preset *. Next return to the configuration menu and set the following:
|COMP ||= ||ON|
Later in Dave's manual is a full description of the PC I/O setup. Again briefly, Dave instructs 2400 baud, 8 data bits, 1 stop bit, NO parity. Finally... the cable listed above is a "Y" type having two separate tails. The common end goes to the X-Rite instrument. The "A" tail to a printer or PC and the "B" tail to a modem (391 only).
|* Note: ||Versions v6313 and higher have an I/O preset designed to quickly pre-configure the instrument. The notes above will work in any case... but if you have the X-Read preset you can save a few keystrokes: Use X-Read rather than Epson where possible.|