Most settings for the instrument are set within the ShadeVision application. To modify your instrument's settings proceed as follows:
From the "Advanced Menu", click on "Instrument Health"
Use the Contrast Slider to adjust the display of the touch pad screen.
Move the slider to the left to darken the screen; to the right to brighten.
To save the contrast setting you selected, click on "Save".
To reset the screen contrast to the factory default, click "Default".
To calibrate your touch pad screen, click on "Start" in the "Touch Screen Calibration" field.
Follow the instructions on your instrument screen. They will lead you through the process of calibrating your touch pad screen.
On the right of the screen, you can see details about your instrument.
The serial numbers of the instrument and the calibration plaque are listed, as well as the model and firmware version