Under some conditions it becomes necessary to reset the microprocessor in an X-Rite Multi-Angle spectrophotometer. Be aware however that an instrument reset is a 'last-resort' decision. Any stored standards or sample data will be deleted if you continue! The following procedure will return your MA (multi-angle) unit factory defaults:
Remove external power from the unit.
Wake the unit by pressing the on/off button.
Press and hold keys 1, 2, and 4 while at the same time pressing the on/off button at the back of the instrument.
Wait about 10 seconds.
The multi-angle instrument will then display "MEMORY INITIALIZED BY OPERATOR Press Any Key".
Only for the MA68 press the left key under "CONT".
Questions? Need a Quote? Contact Us. 1.800.248.9748