Reproducibility - iCPlate2
Q - Why do I get inconsistent reproducibility when I execute multiple measurements with the iCPlate2?
A - Following are some possible reasons for inconsistent repeatability:
1) The configuration of the iCPlate2 is not correct. Please check the settings of the iCPlate2.
2) The iCPlate2 has been moved between the measurements.
3) The device has not been placed on a stable, flat surface.
4) The calibration is out of tolerance. Check the measurement accuracy with the iCPlate2 Target software.
Questions? Need a Quote? Contact Us. 1.800.248.9748