If a SpectroEye instrument is used with Color iQC or Color iMatch software, there is a condition which can cause incorrect readings to be transmitted to the software. The condition which will cause this to happen occurs if the user has used the SpectroEye instrument standalone (not connected to the software) and done measurements of:
Any other mode which uses measurements relative to the substrate
To correct this issue, simply switch the SpectroEye instrument to CIELab or any other absolute color space.
Questions? Need a Quote? Contact Us. 1.800.248.9748